Monitor Ferment from Refractometer Readings
Details: Monitor the progress of a ferment without having to take large samples and use a hydrometer, simply take a small refractometer sample. Entering the initial °Brix reading (pre-ferment) and the current reading will give is all that is required.
Important: There are a lot of approximations involved in this calculator. While this method is extremely useful for monitoring ferments, and on the whole quite accurate, it is not perfect - for example, do not expect it to show a true °Brix of exactly zero when the fermentation has finished.
A set of calculations are performed, first to calculate the initial gravity from the inital brix:
ig = 1.000898 + 0.003859118 * ib + 0.00001370735 * ib2 + 0.00000003742517 * ib3
And then to calculate the current gravity from the initial and current brix:
cg = 1.001843 - 0.002318474 *ib - 0.000007775 *ib2 - 0.000000034 * ib3 + 0.00574 *cb + 0.00003344 * cb2 + 0.000000086 *cb3
The alcohol by volume is calculated thus:
abv = 0.93 * ( ( 1017.5596 - ( 277.4 * cg ) + ( 1.33302 + 0.001427193 * cb + 0.000005791157 * cb2 ) * ( ( 937.8135 * ( 1.33302 + 0.001427193 * cb + 0.000005791157 * cb2 ) ) - 1805.1228 ) ) * ( cg / 0.794 ) )
abv = alcohol by volume
cg = current specific gravity
cb = current Brix reading (refractometer)
NOTE The 0.93 conversion factor was added based on experimental results to make the alcohol prediction for this particular calculator more accurate.
The residual sugar (in grams per litre) is calculated thus:
residual sugar = specific gravity * true brix
Spirit indication is calculated form the current alcohol thus, and used to adjust the current gravity, so that true Brix and residual sugar can be established:
si = (2 * SQRT ( 626159497 ) * SQRT ( 35209254016727200 * abv + 448667639342033000 ) - 33520822512398 ) / 841662180975
This is then used to calculate the corrected SG:
corrected_sg = current_sg - ( 1 - ( spirit_indication / 1000) ) + 1
Note that this alcohol calculation is different than the one used in the alcohol by refractometer and hydrometer as it gives more reliable results when coupled with the other calculations used to monitor fermentation using a refractometer.
VinoCalc by Jonathan Musther - email@example.com